Model: A. Manufacturer: Fluke. About: We are NES Sales, Northeast’s industrial source. Notes: Does not include stand. Sold as-is, as pictured. | eBay!. Fluke A bench multimeter. «on: November 04, , am». I got this oldie from ebay the other day, the seller sold it as untested but it works. Fluke will not be liable for any claims, losses or damages of any kind incurred by any user arising from use of this manual. A Digital Multimeter Instruction.
|Published (Last):||26 May 2005|
|PDF File Size:||16.94 Mb|
|ePub File Size:||2.73 Mb|
|Price:||Free* [*Free Regsitration Required]|
Theory of Operation To obtain a quotation to repair shipment damage, contact the nearest Fluke Technical Center. Each range and function of the A is equipped with input overload protection.
The numerical value of each digit of the instrument display is presented in a four-bit bed format at the DOU output connector. There are exceptions, however.
Rent Fluke A Digital Multimeter 4 1/2 Digit | Transcat
At the end of the integrate period Q14 is turned off, U4 and U5 no longer charge C28, and the charge on C28 is held. Knowing that there is a problem. A separate storage compartment is provided for test leads, power cord, and other compact accessories.
In such case, an estimate will be submitted before work is flukr, if requested.
Aereo Bogota DE Tel: Learn More – opens in a new window or tab Any international shipping and import charges are paid in part to Pitney Bowes Inc. The dc voltage at the input of the A-D Con- verter is integrated charges a capacitor for a controlled amount of time ms. To check a questionable range, select the range manually and check the inputs and outputs given in Table that affect that range.
Select the 20,or V range. The troubleshooting techniques given below should be completed as the first step toward repairing any apparent malfunction in the unit. Strobe signals are applied through U16 and the range switches to the a and 3 inputs. Conductive Pkg Containers Inc. Turn the instrument on and allow it to warm-up for a minimum of 15 minutes. To ensure prompt and efficient handling of your order, include the following information.
The accessories are listed and described in Table Associated Stores Click Here. Connect equipment as shown in Figure 1and perform the following steps. Using the dc current calibrator check Table 4- 1 for correct calibrator sequentially apply the values of dc current indicated in Table 44 to the A 88600a input terminals and select the ranges prescribed.
The bed information presented to the instrument display polarity, overload, range code, and digit flukke also applied to the DOU input. Q3 conducts an amount of current such that the voltage drop on R35 equalizes the input voltages on Pins 2 and 3 of U3. The operation of the range control system can be checked by observing the interaction of U8, U9, U16, and the function switches.
Line potential exists on the fuseholder flu,e the instrument is plugged into the line. When all the display decimal points flash on and off, the battery pack should be recharged by switching the POWER switch to OFF and connecting the instrument to the ac power line.
Model A Outline Drawing 2- 1. If the fljke passes this test, the indication is that the fault is in the Input Divider or in the range selection.
By inserting a voltage at TP1 1all input signal conditioners will be by- passed. AC Converter Block Diagram 3- 5. A schematic of the 80K40 probe is shown in Figure If the instrument is damaged in any way, a claim should be filed with the carrier immediately. When reinstalling flume plug-in boards, insure that all connector pins are properly aligned before seating the board.
A busy flag is provided so that the output data may not be used fluk the A is feeding a new input measurement to the DOU. Discrete Semi Conductor Div. You are covered by the eBay Money Back Guarantee if you receive an item that is not as described in the listing.
Fluke 8600A Portable Bench Digital Multimeter 1000v DC
The contaminants on printed circuit boards can first be loosened by spraying them with Freon T. Each instrument manufactured by the John Fluke Mfg. Calibrated to read rms value of a sine wave input. The slope of the output voltage from U5 is propor- tional and opposite in polarity to the level of the dc voltage from the input divider. Paramus, NJ Wabash Inc.
Fluke 8600A Digital Multimeter
U3 has a negative reference voltage applied to its noninverting input and a greater negative voltage applied to its inverting input. Damage may result if alkaline, zinc-carbon, or mercury batteries are charged in the A. The delay allows adjustment of the zero detect level for comparator, U6. An appro- priate read reference is selected in U8. If the bed information at U8 is cor- rect the probable cause of the failure is U The maximum input to the probe is 30V rms or V dc.